Re: WS-3600 update rate
The only way to improve your update rate would be to run wired mode
to the remote thermo.
From the user manual:
15.2 Data Transmission by 433 MHz Signal: Measuring Interval Thermo-
Hygro Sensor: 128 s (at Wind speed ¡Ü 6.2 mph) or 32 s (at Wind speed
> 6.2 mph, or on Wind Gust display); 10 Min. (if the Base Stationdoes not receive any data after 5 successive attempts all outdoor
displays except the rain amount revert to ¡°---¡°)
15.3 Data Transmission by Cable: Measuring Interval Thermo-Hygro
Sensor: 32 s
The only other method would be to hack the electronics in the WS-36XX
remote thermo as was done with the WS-23XX remote thermo. So far, I
have not seen anyone attempt this.
--- In email@example.com, "Tomer Levy"
> I'm using ws-3600 wireless.
> I see that I get update around every 3-5 minutes, while the display
> on the touchscreen updates every 2 minutes or faster, depending on
> wind speed (?)
> I changed the update rate using HWP software to 1 minute (was 5
> minutes) but still I see no change in the update interval I get
> WUHU. HWP is otherwise causing crashes on my system, so I'm notusing
> I can see in the HWP log file that is generated that the update is
> only once every few minutes and not every one minute, also the
> display on WUHU current status is often not up to date.
> My upload interval to WU is set to 300 sec, but from what I
> understood it shouldn't change anything, I'm also not using
> rapidfire, as I don't see the point if I get readsing from the
> station only every few minutes.
> I want to get thereadouts faster obviously, any idea?