Re: Question about reject known bad wind
- Hello again and thank you for the answer ,my meteo station is La
crosse 2300 and my software is WUHU 215
this hapen yesterday the meteo station console wrriten 57 MPH and
this passed to underground ,i was have marked to reject
known bad wind,but not cut it ,and now i have marked
the buton reject all wind spead above 50 MPH .My data to
I hope to help you
--- In email@example.com, "wuhu_software"
> Bad wind speeds have been reported by users over the past couple
> years. I believe 57.04 and 114.1 mph occur on the WS-23XXstations,
> 75 mph is associated with the WS-36XX station.Underground?
> The following wind speeds are rejected ( +/- .5 mph).
> 57.04, 75, 114.1 mph.
> Do you know the exact wind speed that was sent to Weather
> What is your WU station ID and when did it happen?
> --- In firstname.lastname@example.org, "petros0022"
> <petros0022@> wrote:
> > Good morning
> > I like to ask you about reject bad wind ,i have marked to reject
> > bad wind ,but the software it not reject it and the data
> > it say wind 57 miles ,why do not reject it ??,and now i have
> > the buton reject all wind spead above 50 MPH ,I do good for this
> > problem ??
> > Must to doing something else ??
> > Thank you
> > Best regards from Greece
> > Dimitris
> > http://www.geocities.com/meteo19671967