Re: Need help with tweak data feature
- I will try that and see what happens,,,
I got everything else working good ... so hopefully this will work
out and I go from there.
--- In email@example.com, "wuhu_software"
> Hello Miguel,
> From what you have given, I would say that your temperature is
> roughly a couple of degrees cooler than the analysis temperature.
> The way I roughly calculate the tweak is to calculate a percentage.
> Degree Offset / Average Temperature
> So lets say you want to increase by 2 degrees and the average is
> 2 / 70 = 0.0286
> Then, add 1 to that value,
> Enter that value in the temperature tweak setting and check it in
> couple of days.for
> You can never really eliminate the error, the best you can hope
> is a happy medium.this
> --- In firstname.lastname@example.org, "w4mod" <w4mod@y...>
> > ok, this is my problem, on the report that I get every night
> > comes up:This
> > 24 hours Day time Night time
> > Average temperature error -2.8 °F -4.2 °F -1.9 °F
> > Error standard deviation 1.9 °F 2.1 °F 1.1 °F
> > Worst average temperature error -4.8 °F -6.5 °F -2.7 °F
> > Worst standard deviation 3.6 °F 2.3 °F 1.8 °F
> > NOTE: If the error above is POSITIVE, then it means that the
> > analysis temperature is HIGHER than the reported temperature.
> > means that your sensor is reading COLDER than expected.
> > So, I want to correct that, everything else is working good, but
> > actual temperature,, how much should I tweak the data??
> > Thanks
> > Miguel, w4mod