Re: Reset Question- Rainfall Total
- You must inform the software when you are reseting the rain total. If you are running the latest version, you can hit the rainfall total button on the main screen. This will not only reset the total in the console it in sync the software at the same time.
Since you have already reset the console but did not sync the software you need to:
1) Enter tweak screen.
2) Check Disable all rainfall reporting.
3) Click Save Settings.
4) Wait for a new update from Console.
5) Re-enter Tweak screen.
6) Un-check disable all rainfall reporting.
7) Click Save Settings.
--- In firstname.lastname@example.org, "BrookK" <babblebck@...> wrote:
> Happy New Year ! I reset my data on the HW console this AM. When I did, the total precipitation reset on the console but when I look at it on the left side of the uploader screen it still shows the old amount. This is causing upload errors after 20 trys it stops uploading data....Here is the message
> "Sat Jan 01 13:23:38 2011 > Rain Total reading of 0.77 Inches detected when the average rain Total is 39.87 Inches (over past 10 minutes). Did not replace suspect reading because this has happened more than 20 times."
> I have a Lacrosse ws-2316
> Any ideas what is going on ?