Re: B117, Davis VP & Vue Gust bug fix and other enhancements.
- I appreciate you help as always.
--- In firstname.lastname@example.org, "utahweatherbear" <utahweatherbear@...> wrote:
> no problem...just glad I wasn't imagining the whole thing!
> btw...luv your additions to the log, and uploading 5 minute data to CWOP is awesome as well!
> --- In email@example.com, "wuhu_software" <wuhu_software@> wrote:
> > All,
> > Dan Brown has discovered a bug in the Davis VP2 and Vue gust reporting.
> > The reported gust is 1.6 times too high (it was the metric value
> > reported rather than the English MPH value).
> > Thank you for pointing out this bug Dan!
> > Any Davis user that has installed B115 through B116 should upgrade to
> > B117 to address this problem.
> > For Davis VP2 and Vue users, several new pieces of information can be
> > logged and displayed in the log screen when Show Console Com. Errors is
> > checked.
> > This includes:
> > 1) Console battery level in volts.2) Transmitter low battery warning
> > indicators.3) Transmitter types known and indicator showing which
> > repeater is delivering that transmitter data.4) Various Reception stats
> > about the packets received from the ISS.
> > The CWOP upload interval was added to the CWOP setup screen. This can be
> > changed via the user interface rather than the INI file. You can change
> > this setting only if you are stopped. I believe the minimum value is 5
> > mins and the default is 10 mins.
> > WUHU216_32.zip
> > <http://home.comcast.net/~wuhu_software/files/WUHU216_32.zip>