Re: Thermometer calibration
- Thanks Rick for the suggestion! I'm going to remove the cover today
and see how much that is influencing the temp. If it synchs up more
with the other thermometer, I'll drill some holes in the cover for
ventilation, put it back on, and use the TEAKS for any fine tuning
Given the wiring constraints, the location on the East side of the
house was the best available. It is away from the 1st story roof (to
prevent radiating heat), away from direct sun for all but about 2
hours of the day (8am-10am), away from the direction of heavy rain,
but right on the Southeast corner of the house so that it gets plenty
of wind circulation and no trapped air under the eave. I did think
about the siding, but that's about all that may be affecting it.
It's vinyl, so it cools quickly in the shade - which it is in for all
but 2 hours of the day. Still, I hung the other thermometer right
next to the new one so it would be affected by all of the same
elements and it consistently reads lower.
--- In email@example.com, "Rick" <cfr301@...> wrote:
> There are TWEAKS in the WUHU console to allow for temp errors even
> timed tweaks can be used, open the WUHU program on your screen and
> click on TWEAK DATA. You can actually near the bottom of that
> tweak the temp settings for 2 different time ranges during the day.
> Having said that.. I question placing the sensor where you did,
> because even after the sun is no longer directly on it you may still
> be picking up radation heating from whatever type of siding you
> and air may not circulate as well where you currently have the
> There are many heat shields in the photos section of this group
> along with locations and so on to set your stations sensors.
> Hope this helps
> --- In firstname.lastname@example.org, "ripvanbrown"
> <rascubalon@> wrote:
> > I recently received a LaCrosse WS-2316 Weather Station and
> > the Thermo-Hygro sensor on the side of the house (2nd story)
> > 18" under the eave. Aside from a 2-hour window in the morning
> > it gets direct sun (soon to be remidied), the main problem I
> > have is whether the temperature gauge is reading correctly (for
> > rest of the day). I have another Oregon Scientific gauge that I
> > stationed directly next to the LaCrosse gauge and the LaCross
> > consistently reads 1.2 - 2.3 degrees warmer than the OS gauge. I
> > take the measurements at various hours throughout the day outside
> > that 2-hour window of direct sun. Interestingly, the 2 base units
> > sitting side by side indoors are usually +/- 0.5 but have been
> > much as 1.7 degrees. Questions...
> > 1) Is there anyway to calibrate a thermometer (aside from 32F or
> > 212F)?
> > 2) What is an acceptable range of accuracy on home weather
> > instruments? I would think that +/- 0.5 would be acceptable, but
> > 2 seems too far. It seems like one of the thermometers is off
> > a reasonable toleration.
> > Thanks in advance,
> > Robert