Re: [wsjtgroup] power supply help - RM35
- Hi Claire - correct way to calibrate the meter is to adjust R104 (follow the wire on the + terminal on the meter to find it) with a calibrated meter on the output terminals. As you turn R104, the panel meter reading will change. Before you do this, go back and adjust the screw on the meter face so the meter reads "0" when the power is off.
73 jay k2ovs