Re: T2 and Davis Vantage Pro 2
- I lowered the serial I/O speed and the weather packets started appearing. Thanks.
--- In firstname.lastname@example.org, Scott Miller <scott@...> wrote:
> A '0' beacon rate means don't beacon, so that's working as designed.
> The easiest way to troubleshoot is to use the serial splitter cable, put
> the station on port B, and use the WXINFO command to check that it's
> getting data. You can also use SNOOP to watch port B activity, but make
> sure port A is faster than port B. You can use SNOOP HEX to watch
> binary data, but make sure port A is at least twice as fast as port B.
> You might also try running a lower baud rate for the weather station.
> It should handle 19.2k, but I don't remember what speeds I tested it at.
> On 8/7/2011 5:58 AM, wrgiffen wrote:
> > I have a Davis serial data logger I'm trying to get working with a T2. I
> > have the T2 setup with port A set to "Davis" and "19,200 baud." The data
> > logger is set to 19,200 baud. The T2's output goes to a Yaesu handheld.
> > If I set the T2 to beacon every few minutes I get good beacon packets,
> > but no weather information. If I set the T2 beacon rate to "0" the T2
> > never transmits anything.
> > Any thoughts on setup issues?
> > Thanks, Bill, WG0W