66052RE: [ccd-newastro] Re: Sky Quality Meters
- Oct 3, 2009Rainer,
Thanks for that information. What you say makes a lot of sense. I plan to
run it on a separate machine that will also run the lightning detection,
cloud sensor and weather station software for my website so it will always
stay on, but the only relevant readings will be when the roof is off there
From: email@example.com [mailto:firstname.lastname@example.org] On
Behalf Of Rainer
Sent: Saturday, October 03, 2009 11:31 AM
Subject: [ccd-newastro] Re: Sky Quality Meters
The SQM-LE when it is running produces itself some heat so if you want to
use it only when the roof is open I would dare to say that you will not need
an enclosure as the self produced heat will be more then enough to protect
the circuit from getting humid,
BUT, yes but DO NOT TURN IT OFF after closing your roff.
Keep it running and so there is no chance of forming dew inside.
If you run it and then when you close the roff you power it off the SQM-LE,
the humidity content inside will be relatively higher then outside. (WARM
AIR CAN CARRY MORE HUMIDITY THEN COLD AIR) When the SQM-LE then cools down
you will get condensation inside !
It is easy to make a test with the SQM-LE running while your roof is open.
With the software you can monitor the temperature of the SQM-LE and if you
also monitor your ambient temperature via a weatherstation you will surely
also have the temperature of the dew point. In the moment the SQM-LE
temperature does get the same as the Dew Point then you will start to be
--- In email@example.com <mailto:ccd-newastro%40yahoogroups.com>
, "Christopher J Abissi" <cabissi@...> wrote:
> Thanks, Rainer.
- << Previous post in topic Next post in topic >>