question about safety testing
- --- In firstname.lastname@example.org, "Joerg Hansmann" <jhansmann@g...>
if i understand correctly you have to test every unit that comes out
of the factory with tests according to the iec601 ?
one thing about irda versus cable -
we can use a very simple not irda - infrared comm , that we'll
specially design with full duplex , 115000 baud , and because we have
to build reciever card - we can build it to connect to printer port
throught long 5meter cable . this could be very simple little card ,
that can be put near the device , and he'll get he's power from
the printer port .
i think this interface design could be very simple one .
it's seems o.k. with you ?
thanks yaniv v
> ----- Original Message -----
> From: <yaniv_vi@y...>
> To: <email@example.com>
> Sent: Saturday, December 23, 2000 1:53 PM
> Subject: [buildcheapeeg] summary of iec 601-1
> > hi
> > here's the summary of the standard iec 601-1 for safty of
> > medical devices .optocouplers) would be tested.
> > http://www.bapcoinfo.com/Articles/iec601safetytestingarticle.htm
> Thanks. This is very informative.
> Basically it confirms the 10uA limit for any electrode.
> Further it specifies the voltage at which the insulation (e.g.
>strength of insulation. Basically you apply a high voltage, as much as
> "Insulation breakdown
> IEC601-1 Clause 20
> This test is actually a series of tests designed to test the
> 5500 volts, to an insulation barrier to determine if itwill "withstand" the voltage and not short across.