RE: 1960 Temperature Gauge
- Nov 8, 2013 Expand MessagesView SourceGood advice.
---In Imperial-Club@yahoogroups.com, <randalpark@...> wrote:Voltage limiter is in the gauge. Your replacement has a bad limiter,
hence the pegging. Disconnect the power and refer to OIC web site to
adapt new limiter before reconnecting gauges.