20864Re: [GenesisRadio] Problems with USB?
- Aug 15, 2014I do see the green USB button in GSDR, and see the USB LED on the G59 board light...
Firmware entry OK 1.1.0 version 1.0.1 just the serial nuber is showing up as 003 instead of 30
No noise/signal indicated in the scope, no audio ... I went through the set up and tried the delta 44 as a input and the onboard realtek audio chip as the output... Mixer is the appropriate delta 44 mixer "M-Audio Delta ASIO"
I have a flex 1000 so I have used pwr sdr abit ... I cant tune outside the initial spread of frequencies "96KHz"
i hope that helps this stuff can be hard to explain... :)... I sure you know what I mean but you get lots of practice..
On 8/15/2014 10:29 PM, 'Bruce KF1Z' kf1z@... [GenesisRadio] wrote:IF you see the green USB button in GSDR, and see the USB LED on the G59 board light...then there is no USB problem.What firmware doe the "about" tab show?You have to explain these statements... I have no clue what you mean."Also does not receive a signal or display one in the scope window and will not let me move up and down outside of the set frequency"bruce
stage 5 is completed! after scrounging up a second 220 ohm 1/4 watt resistor out of the Junk box... (measured exactly 220 ohms older carbon /tan style) and installing it as R9R... All the voltages check out and when I bring up the Genesis software if connects to the USB OK.. But if I click on about ... It reports the serial number as 3 instead of 30.. I then tried to reload the firmware and reset the serial number to 30 seen to work ... if I ask it to read the serial number it displays 30... But if you go back to the Genesis SDR to about it reports sn 3.
Also does not receive a signal or display one in the scope window and will not let me move up and down outside of the set frequency... No sure itf this means much but the earlier pre-amp voltage measurement was about 2 volts low ... but test seem to behave like the description of the test..
Before the receiver was built it was fine I wounder if I corrupted the firmware in stage 4 but I don't recall seeing any errors till after the receiver was built.
I tried looking at the connection on the computer plunging it in and out different ports and noticed some garbage in the report of the connection.. using USB view ie
Device Descriptor:bcdUSB: 0x0110bDeviceClass: 0x00bDeviceSubClass: 0x00bDeviceProtocol: 0x00bMaxPacketSize0: 0x40 (64)idVendor: 0xFFFEidProduct: 0x1970bcdDevice: 0x0030iManufacturer: 0x010x0409: "YT7PWR"iProduct: 0x020x0409: "Genesis G59f"iSerialNumber: 0x030x0409: "0030ÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿÿ"bNumConfigurations: 0x01
Not sure what that means I did see some different garbage on the same line on occasion but the string of "Y" shows up and you notice it reports line 30.. I suspect this is why sn 3 may be showing up instead of 30..
My counter works fine on the latter stages for measuring the frequency..as described in a earlier email I have never been able to get a measurement.. I thought perhaps the amplitude of the clock was too weak for my Fluke 1910A to detect it.
In K5RD doc he mentions something about losing the USB communication and replacing C1P and C2p (22pf) with 15 pf caps to get oscillation of of the 16 MHz clock.
Should I try this.. or do you think its something else?
Any suggestions ? I really appreciate any help you can give me...
- << Previous post in topic Next post in topic >>