I am a long time (suffering) 2310 user (3 years now) and have been through
the mill with this product. After countless faults and errors I have just
discovered a new one...my anenometer will display OFL on the display when it
detects wind from the NW or NNW. When the anenometer changes wind direction,
the OFL message disapears. I have changed all of my cables to Cat5E and have
not had any false data for around 6 months. My system operates in full
cabled mode all the time. I use a dedicated P4 2.8Ghz for my weather Pc and
run Weather Display software.
If anyone has any ideas as to what can be causing the latest failure, I 'd
really appreciate the advice.
[Non-text portions of this message have been removed]